DC display power meter
In addition to displaying the parameters of voltage, current, power consumption and energy consumption (KWh) with very high accuracy, the DC display power meter is able to store energy consumption in KWh. In case of power failure, energy consumption (KWh) is stored and will not be reset. In addition, the user is able to define the upper and lower limits of the voltage so that if the device increases or decreases more than the set value, the device will announce an error by flashing the backlight.
Technical specifications of DC monitor power meter
Voltage measurement range: 100 – 6.5 volts
Voltage display format: 99.9-6.5V
Current measurement range: Type 20 (0 to 20 amps without shunt) / Type 100 (0 to 100 amps with shunt)
Current display format: Type 100 (99.99 – 0 amps) / Type 20 (20.00-00.00 amps)
Power measurement range: Type 100 (0-10 KW) Type 20 (0-0 KW)
Measurement and storage range of energy consumption: 09999 – 0 KWh
Measurement accuracy: Class 1
Dimensions of the front frame: length 89.6 mm, width 49.6 mm
Dimensions of the rear frame: length 84.6 mm, width 44.6 mm
Depth: 24.4 mm
How to apply the settings
Turn the backlight on or off
Press the settings button momentarily.
Adjust the upper and lower voltage limits
Hold the Settings button for 3 seconds until the SET phrase appears. Release the button as soon as the corresponding phrase appears. When the three-digit number corresponding to the upper limit flashes, each time you press the button, increase the first digit by one to reach the desired value. Then release the button and wait for the second digit to flash. After the second digit flashes, adjust the second digit according to the previous procedure. The third digit can be adjusted in the same way. After adjusting all the digits, wait for the number on the right, which is the lower limit of the voltage, to flash. Adjust the lower voltage limit according to the procedure. To save the settings, hold down the button for 5 seconds until the first window of the display appears after the pass phrase is displayed.
Reset the amount of energy consumed (KWh)
Hold down the Settings button for 5 seconds until CLR appears. Release the button. When the power consumption number appears and flashes, perform a reset operation with a momentary push and hold the button for 5 seconds to return to the first window if you do not reset.
Adjust current ranges of 50 and 100 amps
These settings are for the 100 amp type. If you connect a 50 amp shunt, set these settings to 50 A and if you connect a 100 A shunt, set these settings to 100 A. The setting is to hold the button for 7 seconds until the CURR statement appears. Then set the desired value by pressing the button once. Then hold down the button for a few seconds until the saved value appears and the first screen appears.
Click to receive advice, price inquiry and order registration